Why do I always receive the error message "THERE IS NO MEASUREMENT DATA FOR OUTPUT" with the ASCII protocol command "Result" when there are valid results during the trigger and all measurements have been selected for output?
|Firmware Version||3.x, 4.x
The error message can occur if you log into the Gocator via the web interface between the Trigger and the Result command. The action of logging into the Gocator via the web interface clears the measurement results.
Please sign in to leave a comment.