- Article Ref
- Date Created
- Fri, 28th Sep 2012
- Date Modified
- Fri, 4th Jul 2014
Gocator showing manual calibrated when I did not use the Calibration option
Why is the Gocator showing "Manually Calibrated" when I did not use the Calibration option?
The resolution that defines the encoder tick/mm conversion is part of the calibration process. When this value is manually entered, the system will show the Manually Calibrated status. Using the Travel Calibration option, the Gocator will perform a travel and alignment calibration and define the resolution (mm/tick) value. The system will now show the Auto-Calibrated status.